for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||6 July 2016|
|PDF File Size:||19.65 Mb|
|ePub File Size:||20.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
A Front-panel Menu Tutorial Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial A front-panel menu tutorial This section is a step-by-step amnual which shows you how to use the front-panel menu. Chapter 1 Quick Start To rack mount the function generator To rack mount a single instrument, order adapter kit Output binary block header WAIT 1!
Used only in the triggered burst mode and triggered sweep mode.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Select the China site in Chinese or English for best site performance. However, you cannot overwrite any of the five built-in waveforms. Decrements Moves the flashing digit to the right The function generator is programmed to output an FM waveform. Sweep Commands see page for more information. Page Index points, number in arb waveform, RS shipping inventory, 15 ports, serial COMarb download example, short form, SCPI commands, power-down recall,baud rate,sinc waveform, power-line voltage cables and adapters, single triggering, fuse, 16 configuration, OFFS commands to select the function, frequency, amplitude, and offset of the carrier waveform.
Then, rotate the handle to the desired position. Page Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator.
If this occurs, select “Save As Chapter 3 Features and Functions Amplitude Modulation AM Modulating Waveform Shape The function generator will accept an internal modulating signal, an external modulating signal, or both. During calibration, output amplitudes are calibrated for both the open-circuit voltage no load and the terminated output voltage loaded.
As you modify the value, the end point is output. For certain applications where you may not want to output the sync signal, you can disable the SYNC terminal.
The number of frequency points in the sweep depends on the sweep time you select and is automatically calculated by the function generator.
Page 90 Chapter 3 Features and Functions Burst Modulation Burst Rate The burst rate defines the frequency at which internally triggered bursts are generated. The function generator outputs one burst or Single initiates one sweep each time you press the key.
The commands used for burst modulation are listed on page The rate is set to baud when the function generator is shipped from the factory. Thanks for the reply.
Select a Web Site
Set the carrier frequency between 10 mHz and 5 MHz kHz for triangle and ramp. However, you cannot use the noise function or dc volts as the carrier waveform. Report item – opens in a new window or tab. This is the correct fuse for all line voltages.
Agilent Technologies 33120A User Manual
Decibels are agilebt ratio or comparison unit and have no absolute meaning without knowledge of a reference or comparison unit. You can configure the function continued generator to automatically recall the power-down state when power is restored.
You can use sine, square, ramp, triangle, or arbitrary waveforms for the carrier waveform. Page 5 Front-Panel Number Entry You can enter numbers from the front-panel using one of three methods. You can select the function, frequency, amplitude, and offset all in one command.
Agilent a output Disable – Discussion Forums – National Instruments
What is the function of TR1 in this circuit 3. Refer to the later sections in this chapter for more complete details on each command.
Output Configuration Chapter 3 Features atilent Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms. Power from the fundamental frequency, and its harmonics, is transferred to spectral components of the rectangular sampling function.
The default amplitude is mVpp into 50 ohms for all functions. Dimension string array 80 elements. The default is BOTH.
Please enter 5 or 9 numbers for the ZIP Code. Be sure to read the security code rules on page before changing the security code. Select a valid country. Agilenr Standard Event enable register is cleared when: