The is a fully programmable triple Output DC Power Supply delivering 0- 30V/A For example: BK PRECISION, , ,V Do not operate the instrument in ways not specified by this manual or by B&K Precision. Failure to comply with these precautions or with warnings elsewhere in . The is a fully programmable triple output DC Power Supply delivering. 0- 30V/A on 2 outputs and V/A on 1 output. Each output is fully floating and.
|Published (Last):||6 April 2006|
|PDF File Size:||7.97 Mb|
|ePub File Size:||11.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
BK Precision Triple Output Programmable DC 30V/3 Amp Power Supply | Instrumentation
Davis eNewsletter Special offers, tech articles, and more! Precisioh 21 Connection diagram example: Page 16 Enter scroll through the complete menu listed below.
Set Current adjust the first channel, press to control the second channel and press to control the third channel.
Tell us how we’re doing Take our survey! Vibration – Leak Detection.
Page 35 the operation condition sub register is reset. After this command executed, the operation condition is reset. By pressing ‘print’ button you will print only current page. When configuring for series or parallel operation, the user must wire the terminals to match the configuration.
ERRor This command is used to obtain error information from the power supply. Water Quality – Analytical.
BK Precision Triple Output DC Power Supply, 2x30V/3A, 5V/3A
Alternatively, press to move the cursor adjust the voltage value Enter using the knob. Return the product in the precisuon packaging to the address below.
Page of 48 Go. Don’t show me this message again. Before test, please select the channel: Press to enter the selected menu function, press to return to the previous menu.
General information The following safety precautions preecision be observed before using this product and any associated instrumentations. Chart Description Time 1 Power mode: Save And Open Configure the software operating parameters and serial communication parameters.
B&K Precision 9130
Level Sensors – Controls. This command queries the time of the output-timer for the current channel. Table nk Contents Add to my manuals Add. For example, to set voltage for the first channel: ENABle This command sets the parameter of the quest event enable register. Safety Information Safety information Please review the following safety precautions before operating our equipment. This command sets the new calibration password. This precisipn reads the actual output current of the power supply.
This command returns the number of the selected channel Save My Shopping List You can view your shopping list or continue shopping. Use these tools to define each test step and add new test steps. To Rack Mount The Instrument 7. Right-click in the Program area. The digital display shows voltage and current settings for each channel simultaneously while providing high programming and feedback resolution of 1 mV or 0.
Safety information Please review the following safety precautions before operating our equipment.
Enter text from picture: This chapter describes how to use a computer to control the output of the power supply.
Exit Menu Power Model: Adjust each output independently or combine in series or parallel to obtain higher output and currents. Air Velocity – Manometers.
CH X Lost Calibration Page 29 The Structure of the condition register is as followed: Scpi Command Overview 5.
Comments to this Manuals Your Name. Pumps, Motors and Compressors. The Structure of the condition register is as followed: The front panel keys and the control knob provide a convenient means to adjust the voltage and current of a selected output; Test and Measurement Equipment. AMPLitude] This command sets the current value of the power supply. Tubing, Valves and Hose.