This calibration manual focuses on performance verification and calibration of the Fluke. A and A Digital Multimeters (hereafter referred to as the. The Fluke A & A digit precision multimeters have the accuracy & versatility to handle your most demanding measurements, on the bench or in a. Manual/User Manual on CD-ROM, FVF-BASIC. FlukeView Forms Software Basic Version. Ordering Information. Fluke A. Digit Precision. Multimeter.
|Published (Last):||26 September 2009|
|PDF File Size:||19.56 Mb|
|ePub File Size:||11.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fluke Calibration Silver CarePlans are available for most calibration products.
To store the present Meter configuration in one of the first four memory locations, press the appropriate soft key. Setting The Display Resolution Front-Panel Operation Configuring the Meter for a Measurement The beeper state is stored in non-volatile memory and does not change when the Meter is turned off or after a remote interface reset. Their full complement of measurement functions plus its RS, IEEEand Ethernet Remote Interfaces makes these multimeters ideal candidates for precision manual measurements and use in automated systems.
Petrochemical industry calibration solutions 2. For best accuracy and stable readings, Filter choose a filter based on the lowest frequency to be measured and the performance needed. First verify that flu,e line voltage selection is set to the correct position, and then that the correct fuse for that line voltage is installed.
This alteration should be done by a trained technician at Fluke service center. These triggers are continuous and occur manuaal fast as the configuration will allow. The data is initially stored in internal memory and then transferred to the USB memory device. Connect the Meter to a properly grounded three-prong outlet.
Front-Panel Operation Introduction Introduction The Meter can be controlled either by sending commands through one of its communication interfaces or by manually operating its front panel controls. To start a histogram session: To reveal signal quality issues like drift, intermittants and stability, view data as a real time trend plot or histogram, with the manuak analyze mode. Multimeter Fluke A User Manual pages.
Flkke To Line Power First verify that the line voltage selection is set to the correct position, and then that the correct fuse for that line voltage is installed. One contact is connected to HI or LO input circuits and the other contact is connected to the Sense input circuits.
Testing Continuity Continuity testing determines whether a circuit is intact i. An example of the diode test display is shown below.
Optional Memory in mqnual storage capacities is available from Fluke. All product names are trademarks of their respective companies. After removing the power-up configuration, power the Meter down and back up using the rear-panel power switch to restore factory default settings.
8845A/8846A 6.5 Digit Precision Multimeters
fluk An example of the capacitance display is shown below. To manually stop the measurement collection, press the STOP soft key from the statistics menu.
Page 9 Page Note that this may take several seconds. Warranty support is available only if product is purchased through a Fluke authorized sales flukd or Buyer has paid the applicable international price.
A/A Users Manual
Crest factor0. Carefully unpack the Meter from its shipping 88446a and inspect the contents for damaged or missing items.
Press the soft key again to disable the function. Basic Manuaal dc accuracy of up to 0. Fluke A Manual pages. Read “Maintaining in-situ traceability on the factory floor with process metrology.
With the test leads connected to the mA and Input LO connectors, press the A soft key as shown above, if not already highlighted. Opening the cover of the Meter to make this alteration may void the This unique combination of features and performance makes the A and Manuwl an unbeatable value for a wide variety of applications, including manufacturing test, research and development, and service. Of course, these DMMs are also durable and dependable, features you expect from any Fluke meter.
This chapter explains the function manuao use of the controls and indicators located on the front and rear panels of the Meter. If the readings fill the last filereading storage will stop. Turn the Meter off, unplug the power cord from the Meter, and remove all test leads. The display indicates the next available memory location. Manuql contact your local Fluke Calibration sales representative for details or to request a quote.
Rack Mount Kit, Single. Don’t have an account? This manual also for: For the A, the lower four keys select different functions; see inset. This filter averages readings Understanding Specifications for Precision Multimeters Power and utilities industry calibration solutions 5.